FORTIN BAROMETER PDF
This instrument is made for measuring atmospheric pressure with the scale in hPa and the venier which can read to 1/10 mm. Since this precision fortin. INTRODUCTION. The Fortin barometer is the most widely used type and can be found in many industrial laboratories, workshops, classrooms and. A wooden wall plaque supports a brass outer casing which protects the glass barometer tube and wood and leather cistern. A brass screw provides the means .
|Published (Last):||28 May 2004|
|PDF File Size:||1.36 Mb|
|ePub File Size:||17.48 Mb|
|Price:||Free* [*Free Regsitration Required]|
It should then be washed and dried. An alternative is to mount the thermometer in a short tube of mercury having the same cross-sectional area as the barometer’s tube, and shielded with the same materials as the barometer’s frame. Simple device used for filling barometers using a vacuum pump. Before travelling, the base of the reservoir is raised using the adjusting screw until both reservoir and tube are filled with mercury.
Measuring temperature Mercury barometers have a temperature coefficient of nearly 0.
Fortin Mercury Barometer – Physics Museum – The University of Queensland, Australia
In the Fortin barometer the bottom of the reservoir is flexible originally it was made from leather and can be raised or lowered by means of a screw, allowing the surface of the mercury to be adjusted to a predetermined level. The level in the cistern is adjusted by adding or removing mercury using a syringe so that the screw W enables the mercury level forhin reach the tip of the ivory pin whatever the prevailing atmospheric pressure.
The pressure in the bzrometer line is monitored using a Pirani gauge and should be no greater than 0. Fortin barometers should be used with their axis of rotation vertical – such that a setting of the mercury surface to the fiducial point remains correct after axial rotation through any angle. The seal between the cistern glass and the flanges is provided by leather washers. The barometer is once again mounted on its gimbal support.
When the clamps are in place, the three long bolts P are slackened and removed. For example, an uninsulated thermometer hanging in air near the barometer will probably be warmed much faster by the presence of the user than will the barometer – leading to temperature and hence pressure measurement errors.
The level of mercury in the cistern is topped up and the process repeated until the mercury will no longer rise further in the tube.
Joint Ventures NPL collaborates with leading national and international organisations to ensure our science achieves the highest possible impact.
A glass tube T, about 10mm OD and 6mm ID, is sealed at the upper end and is provided with a tapered lower end which projects into the the cistern C.
The cistern after removal of the mercury, the barometer tube and the glass cistern tube. Users without a subscription are not able to see the full content.
The Pattern Greensand Casting: After tapping the barometer, the level-adjusting screw S should be turned slowly until the thin background of light between braometer mercury surface and bzrometer point just disappears, while viewed along a horizontal line of sight. Experienced consultants and project managers draw upon a unique combination of industry know-how and world-leading scientific discovery to deliver real-world business solutions and enable innovation and secure competitive advantage.
Alternatively, a thin teflon tube is threaded up barpmeter barometer tube which is mounted in its normal position and the air pumped out through the trap pictured on the left.
One such is to seal a tube on to the normally closed end of the barometer tube, pump it down in situ, thus drawing up the mercury from the cistern. The stopcock on the trap is opened momentarily to remove some air from the barometer tube.
But there is no obligation – please varometer yourself of the information and facilities of the site at no charge. The barometer is designed to read correctly at 0 deg C.
Air pressure on the free surface of the mercury in the cistern supports the column of mercury, the length of which is read fortun a vernier operated by a rack and pinion over a graduated scale marked on the casing. Mercury barometers must be firmly supported, made vertical and given time to adopt ambient temperature before attempting to use them.
Ecology and Conservation GO. Recently viewed 1 Fortin barometer.
A Zirconium Oxide Detector for Oxygen. If g is the local gravitational acceleration and g 0 is the standard to which the barometer is calibrated frequently At the upper end of the instrument, the barometer tube T passes into the brass scale housing D. The stopcock is then opened and the barometer pumped down for an hour. Normally distillation is not necessary. The emptying of the barometer tube can be accomplished by carefully removing some mercury from the cistern using a hypodermic syringe, and then lowering the mercury level using screw W to a point just below the opening of the barometer tube T.
The first step is to dismantle the barometer. A wooden wall plaque supports a brass outer casing which protects the glass barometer tube and wood and leather cistern. The trap fills with mercury during this process. Fortin barometer in A Dictionary of Ecology.